It’s just a simple: The problem I am facing is this. Maintain a constant drinkable temperature for a cup of coffee? I have attched the IV curve im getting thus. I scanned through them, and most of them were too complicated for me to undestand. From there decide which one makes sense based on the instrument’s front panel. I need to make interface so that i can collect data through simple commands.
|Date Added:||17 May 2006|
|File Size:||8.27 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
My question is similar to this man’s question, albeit his program seems a bit too complicated for me to understand.
What interface do you want. Trying to make IV curve. Anyway I now that I’m back, here’s what I managed so far:. Message 10 of There are only two ways to tell somebody thanks: Message 4 of 5. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Message 6 of MAX was able to detect the Sourcemeter. The link to the reference manual did not work for me but I have found a copy of the A reference manual here:.
Write Voltage, Read Current. It does not take the remaining digits.
Facilitate inheritance of any offspring? I have one gpib cable. This is a 2 channel meter, and my plan is to sweep one channel from 0 to 5 volts and measure the voltage it is sourcing, while the other channel is swept from mV to mV so no change in voltaage and measure the keithhley coming through.
Search Instrument Driver Network
Message 2 of 4. I hae used the labview drivers of keithley The issue I am having is with the sync between Channel A and CHannel B, it seems pabview channel is running faster than the other even though there is no delay in theory. Also, some of the included sample files give the same error.
Message 3 of 5. Labview drivers for the Model can be downloaded from the National Instruments site at http: Message 1 of Message 1 of 3.
Increasing Keithley sample rat – Discussion Forums – National Instruments
Message 2 of 5. I would suggest you start small. Message 8 of Maintain a constant drinkable temperature for a cup of coffee? Try the GPIB example: With the second cable, you then connect the two instruments in series with the PC. I didn’t check your program, but the waitcomplete command could help you.
Keithley SourceMeter – Instrument Driver – National Instruments
Message 8 of The Data Bit on Keithley was 8 bit while the program was defaulted to 7 bit. In order to synchronize Channel A and B, you need to share a clock and a trigger.
The problem I am facing is this. Message 10 of